## Don’t Use Barometric Pressure For Calibration

Take a look at the screenshots above. The one on the left is for Death Valley at 282′ below sea level and the one on the right is Denver, CO at 5,280′ above sea level.

Notice the barometric pressure, they are almost the same

This means that barometric pressure is corrected or “normalized” to sea level so that the weather can easily be compared from one place to another.  In other words, barometric pressure is there for weather forecasting not for calibrating tools or for calculating air density. In fact, If you own a simple dial barometer and you change its elevation you would need to recalibrate it to get it to match the forecast. Take a look at the chart below, you will notice that pressure drops by about 1″ for every 1000′ of elevation change. This is the way altimeters in aircraft work, they really aren’t measuring distance, they are measuring pressure and converting it into altitude.

If it wasn’t for this barometric normalization barometers in Denver would hover around 24″hg rather than the 30″ you will find in the forecast.

Of course, you can correct for this by subtracting from the forecast barometer pressure based on the altitude.

For the Denver reading of 30.24 you would do it like this –

5,280 ÷ 1000 = 5.28″ of mercury pressure below sea level

30.24 – 5.28 = 24.96″hg (inches of mercury)

If you want to covert that to PSIA you divide by 2.036 to get PSI

24.96 ÷ 2.036 = 12.26 PSIA

Obviously, none of this is a problem when using tools that you simply “zero” to the atmospheric pressure, but in some cases, an instrument or calculation may require you to enter the pressure directly. This is when using barometric pressure can be an issue… unless you live at sea level like I do.

— Bryan

This site uses Akismet to reduce spam. Learn how your comment data is processed.

Scroll to top
Translate »

### Daily Tech Tip

Get the (near) daily Tech Tip email right in your inbox!